Principle Of Geiger-Muller Counter
Geiger-Muller Counter Construction
Working of Geiger-Muller Counter
The pulse size is the same, no matter what type of energy radiation caused the ionization. For example, a small beta particle would produce a much smaller initial effect than a large alpha, but the avalanche would become so large that it would be impossible to tell what started it. Therefore, in the Geiger-Muller region, it is possible to tell that radiation is present, but it is not possible to determine the type of radiation.
The time during which the second pulse could not be detected IS called dead time and IS equal to approximately 10/1000000 seconds.
The dead time and the resolving time are really dependent on the time it takes for the positively charged cloud of ions to move toward the chamber walls. As we know that, after the avalanching occurs, there are two smaller voltage potentials present, like two sub-detectors, both operating well below the GM regiOn.
produced 'will be very small in comparison with the millions of ion pairs already produced. Thus, the new pulse would go undetected, and this would be the dead time. As the positively charged cloud moves closer to the chamber walls, one large voltage potential and one smaller voltage potential sub-detector are present.
However, the larger voltage is not yet as large as it should be, so the new pulse is not full-sized, and it will not be detected by the outside electronics. When the positively charged cloud reaches the chamber walls, there is only one large voltage potential, and the detector is once again operating in the Geiger-Muller region.
Again, the time before a full-sized pulse could be produced is the resolving time. Geiger-Muller tubes are the detection instruments used in the Geiger-Muller region. Most of those detectors use argon gas within the chamber as a result of argon gas is well ionized. However, there is a problem with argon gas. When the positive argon ion is collected on the conductor, item its photons within the style of UV.
The photons can undergo a photoelectric effect interaction with the walls of the chamber, producing more electrons and starting the avalanche all over again. To stop the continual avalanche, another gas, called a quenching gas, is mixed with the argon. The quenching gas is usually a halogen ' gas such as bromine or chlorine.
Applications of Geiger-Muller Counter
- It is very useful for counting B-particles. It can also be used for measuring the intensity of y-rays, X-rays, and electrons.
- It is also used to record the events involving high energy particles and cosmic rays.
- It can detect even small activities, i.e. it is sensitive to the production of even a single ion-pair.
- It can detect all types of radiation.
- It has a long insensitive time to allow entry of each particle.